Woven EndoBridge (WEB)
Placed in the aneurysm, the device will modify the blood flow at the level of the neck and induce aneurysmal thrombosis. The WEB shape was designed to treat wide necked aneurysm. The device has been developed progressively from a dual-layer version (WEB DL) to single-layer versions (WEB SL and WEB SLS [single-layer spherical]).
This device does not require long-term antiplatelet use.
For the treatment of both ruptured and unruptured aneurysms. The WEB has received the CE mark and to date has been used to treat a wide variety of more than 1,400 aneurysms in Europe, Latin America and New Zealand. The WEB is not available for sale or use in the United States.
The WEB is a self-expanding, oblate, braided nitinol mesh.
The device is composed of an inner and outer braid held together by proximal, middle, and distal radiopaque markers, creating 2 compartments: 1 distal and 1 proximal. Depending on the device diame- ter, the inner and outer braids are 108 wires or 144 wires. Therefore, blood flow into a WEB-embolized aneurysm initially encounters 2 layers of wires comprising 216 or 288 wires, with the largest interwire distance ranging from 106 to 181 m, respectively, depending on the device size. The WEB implant is deployed—or retrieved before de- tachment—in a manner similar to that in endovascular coil systems, through microcatheters with an internal diameter 0.027 inch. For devices with a diameter of 7 mm, microcatheters with an internal diameter of 0.027 inch are used; and for devices with a diameter 7 mm, microcatheters with an internal diameter 0.032 inch are used. The detachment system is electrothermal and instantaneous. 2).
In a study, there was no difference in the early clinical course between those treated with WEB embolization, coil embolization, or neurosurgical clipping. Since WEB embolization is a valuable treatment alternative to coiling, it seems not justified to exclude this procedure from upcoming clinical SAH trials, yet the clinical long-term outcome, aneurysm occlusion, and retreatment rates have to be analyzed in further studies 3).
It does not immediately secure the aneurysm in most subarachnoid hemorrhage cases. Second, it may not be suitable for embolization of wide-neck aneurysms with an unfavorable aspect ratio. To overcome these limitations, Zanaty et al., used the WEB device in conjunction with stenting and/or coiling.
They presented a technical note with an illustrated case-series, and provide a detailed step-by-step description on how the WEB device can be used in adjunct to coiling and/or stenting to achieve successful angiographic results. Accurate sizing of the WEB device before deployment is critical. Larger case-series are required to further assess the safety and success of these combined techniques 4).